Online Dissolved Ozone Meter T6058
Online dissolved ozone meter is a microprocessor-based water quality online monitoring control instrument.
This instrument is widely used in online monitoring of water supply, tap water, rural drinking water, circulating water, washing film water, disinfectant water, pool water. It continuous monitoring and control water quality disinfection (ozone generator matching) and other industrial processes.
85~265VAC±10%,50±1Hz, power ≤3W;
9~36VDC, power consumption≤3W;
Chlorine dioxide: 0~20mg/L; 0~20ppm;
Online Dissolved Ozone Meter T6558
Trend Chart Display
1.Large display, standard 485 communication, with online and offline alarm, 235*185*120mm meter size, 7.0 inch large screen display.
2.The data curve recording function is installed, the machine replaces the manual meter reading, and the query range is arbitrarily specified, so that the data is no longer lost.
3.Historical curve: The dissolved ozone measurement data can be stored automatically every 5 minutes, and the residual chlorine value can be stored continuously for a month. Provide "history curve" display and "fixed point" query function on the same screen.
4.Built-in various measurement functions, one machine with multiple functions, meeting the requirements of various measurement standards.
5.The design of the whole machine is waterproof and dustproof, and the back cover of the connection terminal is added to extend the service life in harsh environments.
6.Panel/wall/pipe installation, three options are available to meet various industrial site installation requirements.
Electrical connection The connection between the instrument and the sensor: the power supply, output signal, relay alarm contact and the connection between the sensor and the instrument are all inside the instrument. The length of the lead wire for the fixed electrode is usually 5-10 meters, and the corresponding label or color on the sensor Insert the wire into the corresponding terminal inside the instrument and tighten it.
Instrument installation method
|Measurement theory||Potentiometric method|
|Temperature compensation||Manual or automatic|
|Electrode residual signal||<1‰|
|Response time||25°C<60S; 35°C<30S (To attain 90%)|
|Stability||At constant pressure and temperature, the weekly drift<2%F•S;|
|Current output||Two:4~20mA,20~4mA,0~20mA(load resistance<750Ω)|
|Communication output||RS485 MODBUS RTU|
|Relay control set-points||Three:3A 250VAC,3A 30VDC|
|Optional power supply||85~265VAC,9~36VDC,power consumption≤3W|
|Working conditions||No strong magnetic field interference except the geomagnetic field.|
|Installation methods||Wall mounting|
CS6530 Dissolved Ozone Sensor
|Double liquid junction,annular liquid junction|
|PP, Glass, 120mm*Φ12.7mm|
|0 - 5.000 mg/L, 0 - 20.00 mg/L|
|None or Customize NTC10K|
|4 core cable|
|Standard 5m cable, can be extended to 100m|
|Tap water, disinfectant fluid, etc.|